Подборка снимков экрана (скриншотов) программы
Путь: начало » скриншоты » стр. 1
XA-series. Digital subtraction angiography mode (DSA-mode). The mask is slightly offset comparing to the image position. The mask number is switched automatically for every image (on the basis of DICOM-headline information).
Automatic image contrasting region search. The "Region" tool. The result is the overlay equivalent to the "hands-free" tool. Area and perimeter calculation, calculation of HU-overlay value averaged through all the area.
MG series. The mammograms display mode is switched automatically on one of the displays (there is one display in the system).
US series. Several measurements are made in different calibration regions. If the measurements in the low area is parallel to one of the coordinate axis, then the section length is calibrated according to the corresponding coordinate. If the section is inclined, then the length is calculated for two coordinates at one and the same time.
The MR study is opened, two series are opened simultaneously, the screen magnifier mode is activated, and user's panel is displayed. The user's panel (as well as the other tools bars can be set at your convenience and placed at any side of your software interface).
"Histogram" window. HU-values histogram is displayed for the section chosen. The cursor is on the histogram (it is cursored with the PC mouse as you point it on the histogram window). The HU value at the cursor point is displayed in the window: Cursor X=102.844 mm Y = 2009.273. The X-axis values can be calibrated on their length (in mm) - in case the image is calibrated.
"Automatic settings" window. For the main window "CT" setting is applied. For all of CT series the 2x2 splitting will be applied automatically. For all the other series splitting will be on default. At the present time there is 2x2 splitting.